- UL 2580
Relative Standard test requirements
- Test Object: Battery pack and system.
- The test ambient temperature should be above 0 ℃, wind speed is not more than 2.5 km/h, and the SOC requirement of the battery pack is more than 50%.
- In the test, the size of the Flat plate with gasoline should exceed the horizontal size of the test object by 20 cm, no more than 50 cm. The height of Flat plate should no more than 8 cm above the gasoline. The test object should be placed in the middle. The distance between gasoline level and the test object is set to 50 cm. Inject water to the bottom of Flat plate before testing. The external burning diagram is shown in the graph.
|Product Name||External Fire Exposure Test Machine|
|Test Space||W1500 X D2500 X H800mm||W2000 X D3000 X H1000mm|
|Equipment Dimension||W7500 X D3700 X H960mm||W8500 X D4200 X H960mm|
|Movable Rack||W1500 X D3000 mm||W2200 X D3200 mm|
|Max. Load||1000KG (2000KG,3000KG can be customized)|
|Object Holder Moving Speed||6000mm/min (Adjustable)|
|Refractory Rack Moving Speed||6000mm/min (Adjustable)|
|Control System||PLC touch screen close control + computer remote control|
|Ignition Mode||Automatic high pressure pack ignition|
|Distance Between Burning Plate and gasoline||The distance between the gasoline and the test object can be adjusted within the range of 450cm~550mm. The defaulted 500mm; the bottom of the gasoline plate is filled with water.|
|Refractory Brick Size||In accordance with the standard of ECER100, Refractory Brick Size 240mm*120mm*70mm.|
|Gasoline Plate Size||3 gasoline plates-can be changed easily according to the test object size. the height of gasoline plate is no more than 80cm from the gasoline level|
|Power Supply||AC 380V,7.0KW||AC 380V,8.0KW|
|Weight||Approx 3.5 Tons||Approx 4.0 Tons|
|Other requirements||The ignition area of gasoline plate can be adjusted according to requirements.
Automatic gasoline filling.
Electrical ignition according to test requirements.
The video monitoring system, lighting system.