The temperature humidity test chamber indicates the effect of temperature changes on materials and products. The temperature humidity test chamber aids effectively in ensuring positive outcomes in products with strict limits to failures.
The humidity test chambers are developed in different sizes with a series of innovative features from brands like DGBell’s temperature humidity test chamber. Most temperature humidity test chambers are small in size for testing smaller products and materials, while the larger machines are for testing bigger products.
However, when it comes to choosing of temperature humidity test chamber, there are several things to consider, which includes:
The normal relative humidity range of environmental test machines range from 20% RH ~ 98% RH. Considering the humidity range when choosing a humidity test chamber is necessary as most chambers have certain humidity indicators. It’s this indicator from the chamber that indicates the type of test supported by the machine. Most humidity test chambers are perfect in getting results for high wet tests, while some are used for low tests.
Another important thing to worry about is the temperature range which often ranges from -70 ~ 150°C. This can be determined based on the temperature of the products or material that will be tested on the humidity chamber.
Researching and comparing series of humidity test chambers along with its ascertain features is important. During this period, choosing the machine with highly improved performance should be the goal. The performance stats of any environmental test machine is what determines its positivity and accuracy output when testing products and material.
The selection of appropriate environmental test machines is always complex because the cost is among the first things to consider. The humidity test machine varies with each other in terms of efficiency, performance, and features that aids in creating different price rates. Therefore, depending on your budget and the need you have for getting a humidity test machine, the evaluation of machine costs based on the features required is necessary.
Machine control method
The machine control methods for humidity test chambers are divided into two including the constant test chamber and alternating chamber, which is different from each other. Therefore the selection of the humidity test machine depends on the user’s needs along with the type of test machine they can control.
In most chambers, environmental test temperature always changes, making it easy when assigning parameters. Although certain test chambers like the DGBell’s temperature humidity test chamber give users the ability to control the test machine effectively by a certain parameter.
Considering the volume available for testing products and materials Is necessary when choosing a humidity test chamber. As indicated earlier that most humidity test machines are big while some are small. The same fact applies to the amount of test volume a machine might accommodate. Therefore evaluating the products or material to be tested on the machine gives the user the ability to get an extended chamber with accurate volume.
Moreover, the last thing to consider is the cooling method which includes the air and water cooling methods. The choice of the cooling method of the machine should be based on the situation of the environment. Although in most cases users prefer making use of the water-cooled version of the temperature humidity test chamber, as it’s friendly with most conditions.